Published in Proceddings of the 27th Antenna Measurement Techniques Asssociation Meeting: Newport, RI., October 31, 2005.
Operating frequencies in the gigahertz range is creating an increased need for electromagnetic compatibility (EMC) testing. In the United States, FCC regulations require conformance to radiated and conducted emissions specifications. An EMC laboratory was established at Cal Poly San Luis Obispo (screen room, test instrumentation, and software) and an experiment was developed to explore conducted emissions effects. This paper will describe the test configuration, explain the calibration procedure needed to acquire accurate measurements, and illustrate measurement techniques applied to two example systems. In addition, the data collection process is illustrated through software donated by CKC Laboratories (EMC specialists). To verify the functionality of the laboratory and to assess measurement accuracy, two 12V/15W switching power supplies are characterized for conducted emissions performance; one as supplied by the vendor (KGCOMP) and a second unit with the EMC filters removed. The noise spectrum for both units are plotted against frequency and compared to FCC specifications. The unaltered unit is shown to be in compliance, thus verifying the accuracy of the test procedure and instrumentation.
Electrical and Computer Engineering
2005 Maximilian Moy and Dean Arakaki.
Number of Pages
Published by Antenna Measurement Techniques Association/A>.